Dear Label_Name,

Greetings! I hope you are well.

I am writing on behalf of an international publishing house, Science and Education Publishing, USA (http://www.sciepub.com).

Science and Education Publishing (SciEP) is a multidisciplinary open access publisher of Journals covering the fields of Science, Technology and Medicine. All SciEP publications are subject to high-quality peer review, editorial and production processes and are freely available online.

Welcome to the April, 2013 edition of SciEP newsletter, providing monthly summaries of articles on science & education related topics of interest to scientist and professionals. The mission of our journals is to provide an open access forum for publishing new findings on issues pertinent to scientific and academic field. We also invite you to submit original research papers, review articles, short reports, or case studies are considered for publication in our Multidisciplinary Journals.

American Journal of Public Health Research
DOI: 10.12691/ajphr-1-1-3

The Implementation of a Tele-Homecare System with Service Oriented Architecture and HL7 Message Transmission Standard
Ching-Sung Wang

American Journal of Medical and Biological Research
DOI: 10.12691/ajmbr-1-1-2

The Influence of Ethanol on Pyruvate Kinases Activity in Vivo, in Vitro, in Silico
Lelevich Sergey Vladimirovich, Khrustalev Vladislav Victorovich, Barkovsky Eugene Victorovich, Shedogubova Tatyana Aleksandrovna

American Journal of Medical Sciences and Medicine
DOI: 10.12691/ajmsm-1-1-1

Clinical and Histological Features in Wegener Granulomatosis
Li Cavoli Gioacchino, Passantino Rita, Bono Luisa, Tortorici Calogera, Ferrantelli Angelo, Giammarresi Carlo, Rotolo Ugo

American Journal of Infectious Diseases and Microbiology
DOI: 10.12691/ajidm-1-1-2

The Curse of Lymphatic Filariasis: Would the Continual Use of Diethylcarbamazine Eliminate this Scourge in Papua New Guinea?
Anne Lanham, Lillian Mwanri

We have released issues of some journals on February 2013.
American Journal of Public Health Research
American Journal of Medical Sciences and Medicine
American Journal of Medical and Biological Research
American Journal of Pharmacological Sciences
American Journal of Clinical Medicine Research
American Journal of Nursing Research
American Journal of Cancer Prevention
American Journal of Medicine Studies
American Journal of Infectious Diseases and Microbiology
American Journal of Cardiovascular Disease Research
American Journal of Biomedical Research
American Journal of Mechanical Engineering
American Journal of Microbiological Research
Journal of Preventive Medicine
American Journal of Food and Nutrition
American Journal of Hypertension Research
American Journal of Sports Science and Medicine
Journal of Cancer Research and Treatment
American Journal of Zoological Research
Journal of Biomedical Engineering and Technology
American Journal of Microbiological Research
Biomedical Science and Engineering

We are pleased to announce that submissions are now being accepted for peer review for all the journals across all our subject areas in SciEP.

Please submit your manuscript via our online submission system:
http://www.sciepub.com/author/signin

To unsubscribe from our mailing list, please click on the following link: unsubscribe

I am looking forward to hearing from you.

Kind regards,

David Rain
Email:join@sciepub.com
Science and Education Publishing (http://www.sciepub.com)

Polyethylene Bearing Stress Fractures after Mobile-bearing Cruciate-retaining Total Knee Arthroplasty: Case Reports

Eshnazarov Kamolhuja Eshnazarovich, Jong Keun Seon*

Department of Orthopedic Surgery, Chonnam National University Hospital, Korea

Abstract: Some reported bearing fractures are due to entrapment of the bearing between the femoral component and posterior edge of the tibial base plate in unstable meniscal –bearing design TKA. However, so far no cases of bearing fracture has been reported for the rotating platform or anteroposterior gliding designs of mobile bearing TKA. The authors describe two cases of polyethylene stress fracture after primary TKA using a cruciate-retaining floating platform mobile-bearing prosthesis (e.motion-FP, B.Braun-Aesculap, Tuttlingen, Germany), and includes a review of the pertinent literature and a discussion of possible fracture mechanisms.

Keywords: TKA, Mobile-bearing, polyethylene, stress fracture

Citation: Eshnazarov Kamolhuja Eshnazarovich, Jong Keun Seon, Polyethylene Bearing Stress Fractures after Mobile-bearing Cruciate-retaining Total Knee Arthroplasty: Case Reports. American Journal of Medical Sciences and Medicine, 4 (2). 23-25. January 2016. doi:10.12691/ajmsm-4-2-1

1. Introduction

Mobile-bearing knees were introduced in the mid to late 1970s in an effort to maximize congruity of the articular bearing surface in total knee arthroplasty (TKA), while allowing tibial motion relative to the polyethylene insert, and thereby, theoretically improve knee flexibility and kinematics. Proponents of this design have indicated that it is more forgiving of surgical technique, and have cited its “self-aligning” characteristics in the axial plane.

Some mobile-bearing TKA retrieval studies based on observations made at revision surgery have shown instability with or without polyethylene insert subluxation, and dislocation [3]. Instability may occur at any time. However, when instability leads to early failure, it can be devastating to the patient and surgeon. One study found that these problems occurred mid-term at 4 and 6 years postoperatively [9], whereas in another, these problems occurred within 6 months of the index procedure [2]. Although many studies have reported bearing dislocations after mobile-bearing TKA, only a few cases of bearing fracture have been reported after TKA, especially after meniscal-bearing LCS TKA.

This report describes two cases of stress polyethylene fracture after primary TKA performed using a cruciate-retaining floating platform mobile-bearing prosthesis (e.motion-FP, B.Braun-Aesculap, Tuttlingen, Germany). In addition, previous reports related to this rare complication are reviewed and the possible mechanisms of fractures are discussed.

2. Case 1

In June 2004, a 65-year-old woman (weight = 58kg, height = 1.43 m, body mass index = 28.4) with a history of symptomatic left knee osteoarthritis underwent TKA using a floating platform mobile-bearing TKA with a 10mm thick polyethylene bearing (e.motion-FP, B.Braun-Aesculap). During the operation, an acceptable joint gap and ligament balance were achieved. No problem was reported during follow-up examinations, and the patient was able to perform high-level activities (she worked as a farmer) for the next five years.

Figure 1. Varus (right) and valgus (left) stress views at 6 years after primary total knee arthroplasty

However, five years after surgery, she felt instability and pain in her left knee when she stood up from a seated position. She had no obvious trauma history. Six months after the episode, she returned to our clinic. A physical examination revealed a clicking sound when the knee was in motion. Anteroposterior and lateral view radiographs of the knee showed no remarkable alteration in polyethylene wear, whereas a valgus stress radiograph showed no space for polyethylene bearing on the lateral side of the knee (Figure 1). We considered subluxation of bearing as a tentative diagnosis.

Figure 2. An intraoperative photographs of superior (top) and inferior (bottom) views of the fractured polyethylene bearting

Figure 3. Replaced tibial polyethylene insert

Revision of the left TKA was performed, and a fracture at the posterolateral edge of the polyethylene bearing (Figure 2) and a small osteolytic lesion on the medial tibial plateau without loosening of any component were noted. Flexion and extension instability were observed for a 10 mm trial insert, but a 14 mm polyethylene trial provided balanced extension and flexion gaps in full extension and in 90° of knee flexion. Therefore, the broken 10-mm tibial bearing was replaced with a 14-mm bearing (Figure 3). At her two-year follow-up the patient had recovered complete function of her left knee, and was free from pain and could walk normally without any support.

3. Case 2

A 59-year-old woman underwent right TKA using mobile-bearing TKA with a 10mm thick polyethylene bearing (e.motion-FP, B.Braun-Aesculap). On a follow-up examination at 2 year after surgery, she showed 0° to 130° of motion and reported no pain. However, at 3 years after knee arthroplasty, she felt a sudden onset, sharp pain in her right knee while walking down stairs. No swelling or knee motion limitation was noted at the time. However, subsequently, she was not able to stand up from a sitting position or walk upstairs without support. She did not recall any history of injury or trauma. CT images showed a posterolateral polyethylene defect and a polyethylene fragment at the supra-patellar pouch. A fracture of the mobile-bearing polyethylene was strongly suspected.

Figure 4. An intraoperative photographs of superior (right) and inferior (left) views of the polyethylene bearing showing the fracture line at posterolateral portion of bearing and failure of the locking ring

Figure 5. Anteroposterior and lateral view radiographs of the knee after polyethylene trial component replace

Revision of the right TKA was performed. A fracture was observed a fracture at the posterolateral edge of the polyethylene bearing (Figure 4). With a 12-mm polyethylene bearing, knee showed balanced extension and flexion gaps in full extension and in 90° of knee flexion, and thus, the 12-mm polyethylene trial component was used to replace (Figure 5). At her one-year follow-up, she had no problems her knee was stable knee, and the patient was pleased with the result.

4. Discussion

Some advantages of mobile bearing TKA like the minimization of wear, improved transfer of torsional stress, and a more forgiving nature regarding tibial component rotational mal-alignment, have been postulated and supported by excellent long-term clinical results in numerous studies. However, several authors have reported complications specific to this design, especially subluxation, and dislocation or breakage of the polyethylene insert [1,2,3,4,5,7,8,9,10].

It is well known that bearing dislocation is more common for mobile-bearing TKA than fixed-bearing TKA. Several factors facilitate dislocation in mobile-bearing TKA, namely, prosthesis design [5,7], progressive instability [7,10], surgical technique issues, or others [6,7,8]. However, only a few cases of bearing fracture have been reported, especially for meniscal-bearing LCS TKA. Weaver et al. [11] reported three cases of lateral bearing fractures in meniscal bearing LCS TKA, and concluded that bearing fractures were due to bearing entrapment between the femoral component and posterior edge of the tibial base plate in unstable TKA. However, so far no cases of bearing fracture have been reported for the rotating platform or anteroposterior gliding designs. Hence, as far as I am aware bearing fractures have not been reported to date for the anteroposterior (AP) gliding design. The e-motion FP type mobile-bearing TKA allows the bearing to glide anteroposteriorly and rotate, as occurs for AP gliding TKA. In our cases, bearing fractures occurred at 5 and 3 years postoperatively without any special trauma history, which suggest that high flexion activity, especially sudden movement of knee, increases posterior bearing translation in unstable TKAs, and that the over-hanging posterolateral portion of the bearing can cause high contact stresses. Consequently, bearing fractures might be occurred due to entrapment of the subluxated bearing between the femoral component and the posterior edge of the lateral tibial runner, as postulated by Weaver et al. [11]. Additionally, during revision surgeries, we had to use thicker inserts of 14mm and 12mm, respectively.

Breakage of a polyethylene insert in modern mobile bearing design is rare. In fact, this is the first report to describe bearing breakage in floating platform mobile-bearing cruciate retaining TKR. In the described cases, subluxation of the insert due to flexion instability is believed to have been responsible for the fractures.

References

[1]   | Bert JM (1996) Delayed failure of meniscal bearing elements in total knee arthroplasty. J Arthroplasty 11(5):611-611.
[2]   | Bert JM (1990) Dislocation/subluxation of meniscal bearing elements after New Jersey low contact stress total knee arthroplasty. Clin Orthop Relat Res 254:211-211.
[3]   | Buchel FF, Pappas MJ (1990) Long-term survivorship analysis of cruciate-sparing versus cruciate-sacrificing knee prostheses using meniscal bearings. Clin Orthop Relat Res 260:162-169.
[4]   | Chang CW, Lai KA, Yang CY et al (2011) Early mechanical complications of a multidirectional mobile-bearing total knee replacement. J Bone Joint Surg Br 93(4):479-483.
[5]   | Hwang BH, Lee WS, Park KK et al (2011) Anterior-posterior glide mobile-bearing total knee arthroplasty complications related to prosthesis design. J Arthroplasty 26(8):1438-1444.
[6]   | Jahng J, Kim HK, Sohn JM et al (2001) Dislocation of mobile meniscal-bearing element by massive joint effusion-a case report. Acta Orthop Scand 72(2):192-194.
[7]   | Kobayashi H, Akamatsu Y, Taki N et al (2011) Spontaneous dislocation of a mobile-bearing polyethylene insert after posterior-stabilized rotating platform total knee arthroplasty: A case report. Knee 18:496-498.
[8]   | Ridgeway S, Moskal JT (2004) Early instability with mobile-bearing total knee arthroplasty. A Series of 25 cases. J Arthroplasty 19(6):686-693.
[9]   | Sa´nchez-Sotelo J, Ordonez JM, Prats SB (1999) Results and complications of the low contact stress knee prosthesis. J Arthroplasty 14(7):815-821.
[10]   | Turki H W, Trick L (2011) Complete 180° rotatory dislocation in a mobile-bearing knee prosthesis. J Arthroplasty 26(4): 666. e1-e3.
[11]   | Weaver JK, Derkash RS, Greenwald AS (1993) Difficulties with bearing dislocation and breakage using a movable bearing total knee replacement system. Clin Orthop Relat Res 290:244-252.